Examination Therapy Equipments
Place of Origin
Beijing, China (Mainland)
Principle of the CMS50I Pulse Oximeter is as follows: Photoelectric Oxyhemoglobin Inspection Technology is adopted in accordance with Capacity Pulse Scanning & Recording Technology,the Pulse Oximeter can be used in measuring the pulse oxygen saturation and pulse rate through finger.The product is suitable for being used in family, hospital, oxygen bar, community healthcare, physical care in sports (It can be used before or after doing sports, and it is not recommended to use the device during the process of having sport) and etc.
Small in volume,light in weight and convenient in carrying
Operation of the product is simple ,low power consumption
Operation menu for the function setting
SpO2 value display
Pulse rate value display, bar graph display
Pulse waveform display
With PI display
The display mode can be changed
Screen brightness can be changed
A pulse rate sound indication
Multi-time segment storage
With measured data overruns limits and low-voltage alarm function,the upper/down alarm range can be adjustable
Battery capacity indication
Low-voltage indication: low-voltage indicator appears before working abnormally which is due to low-voltage ,and with alarm function
With SpO2 value and pulse rate value of storage, the storage data can be uploaded to computers
Real-time data can be transmitted to computers(Only CMS50IW)
Connected with an external oximeter probe
Power-off :long press power button to turn off the device
Wireless communication function(Only CMS50IW)
Display Mode:1.5" Color OLED display
SpO2 Measuring Range:0%~100%, (the resolution is 1%).
Accuracy:70%~100%:±2% ,Below 70% unspecified.
PR Measuring Range:30bpm~250bpm, (the resolution is 1bpm)
Accuracy:±2bpm or ±2% (select larger)
Measurement Performance in Weak Filling Condition:SpO2 and pulse rate can be shown correctly when pulse-filling ratio is 0.4%. SpO2 error is ±4%, pulse rate error is ±2 bpm or ±2% (select larger).
Resistance to surrounding light: The deviation between the value measured in the condition of man-made light or indoor natural light and that of darkroom is less than ±1%.
Power Consumption:less than 100mA
Power Supply:Voltage 3.7 rechargeable lithium battery × 1
Battery working hour:10 hours
Battery working life:Charge and discharge no less than 500 times.
Safety Type:Interior Battery,BF Type
Sell in standard
A user manual
A data line
A disk (PC software)
An adult-oximeter probe
Two One-off adhesive SpO2 probe
A power adapter(GTM41076-0605 or CMS0105
Sell in addition
Other oximeter probe(Refer to probe application instruction for details and notice renewal)
Dimension:63(L) × 55(W) ×15(H) mm
Weight:About 45g (with the lithium battery*1)